EHD (Ultra Thin Film Measurement System)Part code: EHDSYS
Main product features
- Evaluation of film forming oils and greases.
- Starvation and reflow characteristics of grease lubricated components.
- Fuel economy prediction of candidate crankcase oils.
- Performance prediction of oil in water emulsion rolling mill lubricants.
- Fundamental investigations of the high pressure/high shear behaviour of fluids such as liquid crystals.
- Investigation of boundary additive performance.
- Technical Specs
|Film Thickness||1 to 1000nm|
|Load||0 to 50 N|
|Contact Pressure||0 to 0.7 GPa (Steel ball on glass disc)|
|0 to 3 GPa (WC on sapphire)|
|Speeds||0 to 4 m/s|
|Temperature Range||Ambient to 150°C|
|Test Sample Volume||120 ml|
|PC||Custom software running Windows 7 (32 bit)|
|Safety Checks||Dual Platinum RTD's for temperature measurement Emergency stop on front cover|
|Power Supply||100-240V. 50/60 Hz, 750 VA|
|Mechanical Unit||50 x 50 x 30 cm, 25 kg|
|Electronics Unit||50 x 44 x 23 cm, 19 kg|
The EHD2 Ultra Thin Film Measurement System is the latest evolution of the successful EHD instrument. It is now a fully automated computer controlled instrument for measuring the film thickness and traction coefficient (friction coefficient) of lubricants in the elastohydrodynamic (EHD) lubricating regime. The instrument can measure lubricant film thickness down to 1 nm with a precision of +/- 1 nm. The contact pressures and shear rates in this contact are similar to those found in, for example, gears, rolling element bearings and cams. In addition to film thickness measurements, traction coefficients can be measured at any slide/roll ratio from pure rolling up to 100%.
The instrument measures the lubricant film thickness properties in the contact formed between a ¾“ (19.05mm) diameter steel ball and a rotating glass disc by optical interferometry. The lubricant film thickness at any point in the image can be accurately calculated by measuring the wavelength of light at that point. Normally the system measures the wavelength of the light returned from the central plateau of the contact and hence calculates the central film thickness.
The control software runs on a standard PC and allows the user to easily define a test profile containing a sequence of temperatures, loads and speeds. After setting the initial film thickness, the selected profile steps the instrument through the test sequence, recording data as required, without any intervention by the user.
1) The contact is illuminated by a white light source directed down a microscope through a glass disc on to the contact.
2) Part of the light is reflected from the Cr layer and part travels through the SiO2 layer and fluid film and is reflected back from the steel ball.
3) Recombining the two light paths forms an interference image which is passed into a spectrometer and high resolution black and white CCD (charge-coupled device) camera.
4) The camera image is captured by a video frame grabber and analysed by the control software to determine the film thickness.
EHD 3D SLIM
EHD Ball Carriage Assembly
EHD Ball Shaft Oil Seal Assembly
EHD Grease Scoop
EHD Lubricant and Pot Temperature Probes
EHD Film Thickness Discs
- How do we breakdown the stereotypes and increase the attractiveness of STEM?
- 60 Seconds in the Spotlight: Arup Gangopadhyay, Lubrication Science & Mechanical Friction- Technical Leader, Ford Motor Company
- 60 Seconds in the Spotlight: Dr Peter Lee, Chief Tribologist at South West Research Institute
- EHD Brochure